**0.01A to 50Arms continuous****100Arms 10 seconds****DC-100 kHz frequency range****Extend current range of DMMs, voltmeters and wattmeters**

**Use with phase angle meters****Safety type connections****Insulated construction****Traceable to international standards via NIST and UKAS**

The option 02F/B is a an non inductive shunt designed to operate over a wide bandwidth, with both AC and DC power systems, especially where phase shift phase angle is a consideration, for power calibration systems. Very useful for extending the range of Multi-meters and voltmeters. The option 02F/B can be used with phase angle meters, where phase angle between voltage-voltage, voltage-current or current-current i s required, ideal for phase protection, power meters, wattmeters, energy meters and current transformer calibration.

All measurements made using the option 02F/B shunts are traceable to National and International standards; through the measurement standards of Powertek. The option 02F/B shunts are supplied with a certificate of conformance necessary for quality assurance standards such as IEC17025/ISO9001. Independent measurement certification is possible using UKAS, A2LA or Z540/NAVLAP certificate.

**Calibration laboratories****Phase meter****Single phase angle indicators V-A****Three phase phase angle indicators V-A****Three phase angle indicators V-A****Cos Phi indicators****Power factor meters****Watt meters****Multifunction watt transducers****Phase protection relays****Power meters****Energy meters****Current transformer calibration****Current probe phase delay****Transformer phasing****Synchro****Resolver Angle indicators**

Non-inductive current shunt Option 02F/B, used with DMMs, power meters to extend current measurement ranges. Use with phase meters to enable phase angle angle measurements between voltage and current

Option 02F/B current shunt specifications |
|||

Phase |
Accuracy (23º ± 5ºC) | 0.1Hz to 99Hz | ± 0.02º |

100Hz to 999Hz | ± 0.04º ± 0.005º/kHz | ||

1kHz to 99kHz | ± 0.05º ± 0.005º/kHz | ||

Frequency |
Range | DC to 100 kHz depending on ohmic value | |

Amplitude accuracy | ±0.3% (23ºC ± 5ºC) | ||

Inputs |
Current ranges and nominal resistance | 10mA-2Arms (0R50), 10A (0R02), 20A (0R01) and 50Arms (0R001) | |

Bandwidth | Isolation (inputs to ground) | ||

Isolation (inputs to ground) | n/a | ||

Environment |
Temperature | Operating | -20ºC to +70ºC |

Storage | -40ºC to +85ºC | ||

Humidity (non-condensing) | 10% to 90% RH | ||

Dielectric Strength |
Inputs to case | 1.5kV AC 50/60Hz 1 minute | |

Mechanical |
Weight | 0.2kg | |

Dimensions (W x H x D) | 215mm x 144mm x 390mm |